LifeScan Announces Voluntary Recall of All OneTouch® Verio®IQ Blood Glucose Meters
Johnson & Johnson: March 25, 2013
LifeScan, Inc. is initiating a voluntary recall and replacement for all of its OneTouch® Verio®IQ blood glucose meters in the United States, effective immediately.
LifeScan is recalling and replacing all OneTouch® Verio®IQ Meters because at extremely high blood glucose levels of 1024 mg/dL and above, the meter will not provide a warning that the blood glucose is extremely high and will shut off, thereby potentially leading to incorrect treatment and delaying proper treatment. Read more